|Sensor format||1 inch|
|Partial reading mode||Possible to integer just a part of the sensor (ROI) and display only this window on the video output. This option allows a frame rate increase on the ROI|
|Min/Max Settings for display – Histogram Stretching||Automatic/Manual.|
In Automatic the gain and offset are calculating depending of the histogram.
In Manual you can choose the gain and offset you want to apply on the image.
|NbIgnorePixels||In percentage of the total number of pixels in the image. Number of pixels ignored for the histogram calculation.|
|NUC correction||Correction 1-Point or 2-Point calibrated in factory (possible to realize it also by user)|
In manual you can load any NUC calibration file from the flash.
In automatic the camera interpolates the NUC depending of the exposure time.
|Bad pixel correction||Correction of bad pixel calibrated in factory (possible to realize it also by user)|
|Control of the temperature||Single stage TE cooler|
Note: The NUC and BPR files are delivered only for +15°C
|Cooling capacity||ΔT = 30°C|
|AET (automatic exposure time)||On/Off|
The AET calculates automatically the exposure time depending of the histogram Mean.
|AET Setup||The setup is selectable from 0 to 100% related to the entire dynamic in 14 bits (0 to 16383).|
|Trigger delay||Delay selectable|
|Spectral response||0.9 to 1.7um|
|Full well capacity (in CTIA)|
|Output type||14 bits (RAW) or 8 bits (with Min/Max stretching histogram)|
|Frame rate||Up to 225fps full frame|
|Trigger||IN/OUT (LVTTL), selectable delay|
|Temperature Range||-20 to 71°C|