|Sensor format||1 inch|
|Partial reading mode||Possible to integer just a part of the sensor (ROI) and display only this window on the video output. This option allows a frame rate increase on the ROI|
|Min/Max Settings for display – Histogram Stretching||Automatic/Manual.|
In Automatic, the gain and offset are calculating depending on the histogram.
In Manual, you can choose the gain and offset you want to apply to the image.
|NbIgnorePixels||In percentage of the total number of pixels in the image. The number of pixels ignored for the histogram calculation.|
|NUC correction||Correction 1-Point or 2-Point calibrated in the factory (possible to realize it also by user)|
In the manual, you can load any NUC calibration file from the flash.
In automatic, the camera interpolates the NUC depending on the exposure time.
|Bad pixel correction||Correction of bad pixel calibrated in factory (possible to realize it also by user)|
|Control of the temperature||Single-stage TE cooler|
Note: The NUC and BPR files are delivered only for +25°C
|Cooling capacity||The cooling temperature capacity greatly depends on the heat dissipation. For more detailed information, contact us|
|AET (automatic exposure time)||On/Off|
The AET calculates automatically the exposure time depending on the histogram Mean.
|AET Setup||The setup is selectable from 0 to 100% related to the entire dynamic in 14 bits (0 to 16383).|
|Trigger delay||Delay selectable|
|Spectral response||0.9 to 1.7um|
|Full well capacity (in CTIA)|
|Output type||Analog video PAL or NTSC|
|Frame rate||Standard frame rate: 25Hz (PAL), 30Hz(NTSC)|
|Trigger||IN/OUT (LVTTL), selectable delay|
|Temperature Range||-40 to 71°C|